My oscilloscope's inputs are rated "250V DC + AC peak (10 KHz)." Does this mean it would withstand mains voltage (325V peak) or only 250V DC? I have no plans on doing this any time soon, I am only curious.
- Anybody can ask a question
- Anybody can answer
- The best answers are voted up and rise to the top
if you add peak voltage to your dc offset, it must be less than 250V.
this was verified with dc and a 10kHz ac wave.
AC signals sometimes ride on a DC voltage level. On the scope you may see a sin wave of +60 to - 40 VDC. This is 100 V Ac signal riding on a 10 V DC voltage.
I once worked with a 10 V peak to peak sin and cos signal from a resolver. The reference signal had a 0 V, 400 mV or 800 mV DC level applied to it. A multiplexer would switch between 3 different resolvers based on DC level.
Your scope could handle anything from 250 V peak AC signal to a peak 125 VAC on top of 125 V DC to a pure 250 V DC or any mix as long as the AC Peak signal + the DC level is less then 250 V
For dealing with High voltage signals, use high voltage probes. These typically reduce the signal by a factor of 10 or more. So a 250 VAC signal become a 25 V signal.