Do we need to connect different frequency sine wave sources to spectrum analyzer? Or will spectrum analyzer supply different frequencies of sine wave source to the DUT? Could someone explain the basic operation of a spectrum analyzer? Does a spectrum analyzer just measure the "different frequency components" of a 10kHz sine wave source? That means the 10kHz sine wave composed of many different frequencies.
closed as too broad by Matt Young, Chetan Bhargava, tcrosley, Andy aka, placeholder May 11 '14 at 12:32
Please edit the question to limit it to a specific problem with enough detail to identify an adequate answer. Avoid asking multiple distinct questions at once. See the How to Ask page for help clarifying this question. If this question can be reworded to fit the rules in the help center, please edit the question.
There is a large amount of information about how spectrum analysers work on the Internet.
Variations are possible, but a spectrum analyser usually acts like a tunable receiver that "sweeps" across a frequency range of interest and measures the amplitude of signals (or energy in a known impedance load) present.
A spectrum analyser can be combined with a compatible controlled frequency source to test a filter or similar. The frequency source is input to the filter and the analyser measures the output and controls the source so that the source and analyser track. A similar result can be obtained by using a flat wideband noise source as input.
- Agilent / HP intro - 80 pages of slides - gotta be good for you.
- Spectrum Analyzer article at Wikipedia
- YouTube video - Good introduction - use with a function generator.
- Radio electronics tutorial
- Basics - ECE UCSB article (PDF)
- Tektronix (The MastersTM) - 10 short video tutorials
- Anritsu even! - looks good
- Tutorial from TestRF.com
- Uncle Tom Cobbly & all ...