I just got my first multimeter from Home Depot and it was $60. It's the Extech EX330 and it works fine except for when I measure current. I am doing it correctly and the fuse has not been blown. When I put it on the amp setting it reads .06 A when I put it on milli amp setting it reads .06 mA and when I put it on the micro Amp setting it reads .6 uA. I have no idea if I should take it back or if I am just reading it wrong. Please help.
Not to point out the obvious, but reading the manual seems like a good idea:
I know you said you were measuring current correctly, measuring voltage versus current on most multimeters involves moving one of the test leads to a different jack on the meter. Usually current measurements are all done from a jack dedicated to current measurement. However steps 2 through 4 point out that measuring current at different ranges makes use of different ports. In this case, measurements from 0 to 400mA both use the same jack as for voltage measurements (V/mA/μA/CAP/Ω/Hz/Temp). I find this a bit unusual, but it could be somewhat convenient for low-current measurements.
As to why the meter shows the same digit (6) at different ranges, most likely because the meter was trying to read current from the jack which wasn't connected to anything during measurements.
Usually you have to move the test prod to different sockets for current. If it's [this] (http://m.homedepot.com/p/Klein-Tools-Digital-Multimeter-MM1000/202521270) one then there are two positions for current. It's very important that they be returned to the voltage input before you attempt to measure volts.