I have half bridge load cells similar to the ones in the image below. While I understand the circuitry and hooking up the differential output of the load cell to an INA, I am not sure how these are physically mounted to create strain and give a differential output. So far I have used only beam load cells which are easy to mount.
Any help on this would be appreciated.
@Nedd Thank you for the reply. I have the part without the bracket and I was trying to figure out how to 3D print a part that would act as a bracket. Your answer gave me a lot of insight. I guess I need to make sure of the following points
"Inner E" shape does not have a firm support below it and it displaces when a load is placed on the bracket.
The bracket is mounted in such a way that the pressure is applied to far end of the inner E shape.