It's understandable that a CT, needs to be shorted (or a burden resistor) if there's nothing connected to its output, otherwise it would work as a step-up transformer.
As they state in the user manual:
Current measuring input terminals must be connected through a R/C Measuring transformer with one lead connected to Earth. Direct connection to the line voltage is not allowed.
I wonder what is the technical/scientific reason behind this.