I have a High-Frequency inductor available and I want to measure the inductance using the Rohde&Schwarz HM8118 LCR meter. If I change the frequency to 100kHz and voltage level to 1.5V (max) I get a value of 0.5 Henry which is way too high.
User manual of the LCR meter:
The inductor is made of two ferrite cores as shown in for example here:
How can I correctly measure the inductance (should get a value in microhenries) ?
Square wave signal of 100kHz is applied across the inductor (having no DC component). The measured inductance value also has a negative sign in front of it.