On a 3-wire load cell that has a black , red, and white wire......... I would think if I ohmed out the resistance between the red and white or red and black or any combination there of, that I would see some kind of change in resistance when putting as much as 200 lbs load on the load cell.No change in resistance at all, so how does the load cell develop a voltage drop,even in the milli-volt range if there is no change in resistance as weight is put on the cell?
closed as unclear what you're asking by Andy aka, Enric Blanco, laptop2d, Dmitry Grigoryev, Wesley Lee Jun 30 '17 at 19:44
Please clarify your specific problem or add additional details to highlight exactly what you need. As it's currently written, it’s hard to tell exactly what you're asking. See the How to Ask page for help clarifying this question. If this question can be reworded to fit the rules in the help center, please edit the question.
I would expect that you found the largest resistance was red to black, with the white wire being more or less the centre tap of the total resistance.
Load cells are not designed to be measured as resistance. While the resistance does change with load, it is too small to be measured reliably, and certainly too small to be detected above the noise and digit resolution on a standard multimeter.
It is designed to be measured by a Wheatstone Bridge
Your strain gauge comprises the two resistances R1 and R2, one increases, the other decreases with strain. You supply R3 and R4, and they are nominally equal. The output voltage proportional to strain is taken across Vg, with a high gain low drift differential amplifier.
Changes of strain gauge temperature will affect both resistors equally, and will not be measured at Vg. Changes of excitation voltage across A-C will affect the gain of the measurement, but will not add an offset.
You have not shared the details of the load cell so its hard to comment on how much the resistance should change.
But usually, in the data sheet, it is mentioned that what is the minimum change in load that will cause a change in resistance.
And as @brhans mentioned in comment, the resistance is usually too small to measure through your normal multimeter, that's the reason Wheatstone's bridge is used to measure the change.
Probably you can look on this website for more details.