# Accuracy of a single to differential ended circuit

I have single-ended senors/transducers which are hooked up to a DAC in single-ended configuration. The transducers are calibrated by another company with certificates and they can output from -10V to +10V DC.

For mitigating common mode noise issues I want to convert these transducer outputs to differential ended. Currently I'm using coaxial cable with BNC connectors in single-ended earth referenced configuration and the sources/transducers are floating. I want to switch to diff. ended configuration and plan to use twisted shielded cable with BNC connectors. Below illustration roughly shows the current and planned cases:

After some research I found the following topology for single to differential conversion. LT1054 is just there to create negative supply for the opamp rails.

My questions are the following:

1) Now the voltage measured by the DAC which is Vo+ - Vo- is two tiimes the transducer's output voltage. I can see this is expected because of mirrored outputs. In this case should I reduce R3, R4, R5, R6 to 5k or simply divide the voltage by two in software end? Which is more common or convenient? (edit: I forgot to mention the DAC range is -10 to +10V so for 7V transducer output it will be in trouble to measure 14V)

2) My concern is that in this topology one solders the resistors himself. So the resistor tolerances and accuracy would effect the output. I mean if the transducer outputs 1V the diff output is 2V in simulation. That's alright and expected. But in real those resistors together with wires will not be exactly 10k. So instead of 2V it might read 2.1V which is quite large error. So Im wondering: Am I in wrong direction when trying to convert accuracy important single ended sensors to diff. ended myself this way? Or should I use a ready chip? When trying to mitigate common mode noise I dont want to introduce offset error to the transducer output by this circuit.

• You have not written anything about what precision you need, neither what kind of signal (frequency, voltage, etc) you have. Do you care about DC accuracy? Sep 27, 2017 at 13:59
• I dont want to lose more than %0.1 precision or not possible at least %1. DC accuracy important. Most traducers output DC voltage signals like temperature sensors ect. Sometimes strain gauges, load cells ect. Frequencies are less than 20Hz in general. Sep 27, 2017 at 14:05
• 0.1% of what? 1kV? or 1nV? What is the voltage range? And how good do you want your DC accuracy? In percent and absolute? Sep 27, 2017 at 14:07
• input voltage range is -10 to 10V. accuracy I mean ( output voltage-input voltage) / input voltage Sep 27, 2017 at 14:12
• I know what accuracy is. DC accuracy cannot be specified by just a percent number if you include 0V in your voltage range. There is always a DC offset. E.g. the opamp you used in the circuit above has a spec'ed DC offset of almost 1mV. Which gets you 100% error if you are measuring 1mV, but only 0.1% if you are measuring 10V. Sep 27, 2017 at 14:14