I was measuring the current through current mirror, when I experienced a problem. But first, I want to mention that I measured this with two ammeters of same manufacturer and same model (Mustool MT826 Digital Multimeter). Both were set to mA (milli ampere) setting of measurement and through both of them, the current measured was about 0.5 mA for both collector currents through Q1 and Q2. Then I thought that it would be better to set both of meters to uA (micro ampere) setting of measurement, and so I did. But only one at the beginning; AM1 was set to mA setting while AM2 was set to uA setting - there I spotted a problem. Suddenly, the current through AM1 was around 2.5 mA while current through AM2 was decreased to 300 uA.
So this got me thinking for a while and I came to conclusion that at different settings for current measurement, the ammeter is "seen" as different load to the emitter of Q1 and Q2. So, my question is:
- Is it normal for ammeter to change its internal resistance, when we switch between different settings of measurement (uA, mA, A) or is this due to bad measuring equipment bought?