How to extract the switching noise component for this signal data?

A servo accelerometer when powered by batteries has a clean output, but when used with this SMPS supply its output seems like have switching artifacts from the power supply(the datasheet says its switching freq is 100kHz). Now since the desired range is very low I can easily filter out such noise at the sensor output.

But having said that, I want to know what we can say about the noise at the accelerometer output. So by using a digital scope(at AC coupling mode) I made two measurements and logged the time series data for the standing accelerometer output. The first one is for longer period, here is the screenshot:

Then I plotted the time series and FFT of this as follows:

Then I did the same thing for a smaller time scale:

So by having all these information what can we say about the frequency of the switching noise? I'm trying to learn in such a situation how to quantify the switching noise. Should we look at the time series data or the spectrum for that? And for this particular case what steps should one follow?

EDIT:

For a much longer acquisition I observe the following:

MEASURED DIRECTLY AT THE POWER SUPPLY OUTPUT:

Closer view:

And spectrum:

Supply Output at 50us scale:

Time series and FFT:

The PSU noise(yellow) directly appears at the sensor output(blue):

• If that timebase is 250 nanoSeconds per division, then with 2 cycles per division you have 8MHz ringing. And some 2MHz (2 divisions period). To filter that, use 1uHenry and 1uFarad, which resonate at 160KHz, giving 100:1 attenuation to the 2MHz and another 16:1 attenuation to 8MHz.. Commented Sep 25, 2019 at 9:28
• Well just by judging the time between the pulses, they are not with a fixed frequency, there is quite noticeable jitter on it. And by counting the number of events over the time you plotted you get an average frequency of 15-16 kHz. So I'm not really convinced that it is the switching noise of 100 kHz doing that. Commented Sep 25, 2019 at 9:35

If I'm interpreting your first scope plot correctly it's showing switching noise from a SMPS running just under 100KHz. The apparent jitter is due to the fact that noise occurs on both edges of the switching waveform, so analysis of this waveform will show a fundamental at just under 200KHz, spread somewhat by the jitter on the switching edge. You can see this nearly buried in the noise on your first frequency plot.

However, there's very little fundamental in this waveform. Most of the energy is in the ringing, which you can see in the higher order spectral content.

• How did you quantify the switching noise close to 100kHz? Between each spike I see like 50us to 60us that makes below 20kHz isnt it?
– cm64
Commented Sep 25, 2019 at 13:23
• You're right, I misread the scale. I'll give it some thought and come back with an edit. Commented Sep 25, 2019 at 13:53
• Please also see my edit meanwhile.
– cm64
Commented Sep 25, 2019 at 13:58
• I will also add info from directly supply output.
– cm64
Commented Sep 25, 2019 at 14:00
• Can you post the supply output on a 50us scale? Commented Sep 25, 2019 at 15:26