# What's the reference dB level in spectrum analyser

I am using a Spectrum analyser.

Measuring the 248kHz buck switching frequency and its harmonics in my board. I have set the frequency range from 200kHz and 1MHz. Using a near field H-Probe

When I give peak search on the spectrum analyser, I get a peak at the 248kHz freq with amplitude of 40dB and when I change the amplitude level in voltage, I get -14dBmV.

I also get other harmonics too.

I just want to understand two things :

1. I know the dB is just the ratio of two quantities (power or voltage). But when I get 40dB and -14dBmV, what is the reference level that is taken by the spectrum analyser which it compares with the output power? I assume the spectrum analyser takes a reference voltage or power level and compares it with my board output level and then only it provides the value of 40dB and -14dBmV. Am I correct?

2. What does the +ve and -ve dB value signify in electrical terms?

Please understand that I am just starting the learn the uses and the operation of spectrum analyser. Would be glad if someone could help with the answers in simpler terms.

Thank you

Edit :

Please find the attached image from this AppNote from Keysight

The images shows a reference level set at the top. How is this value chosen and on what basis?

• "dBmV" means the reference level is 1 mV (rms, I believe). – The Photon Nov 13 '19 at 3:10
• Can you share a screen shot showing where you're seeing "dB" and where "dBmV"? – The Photon Nov 13 '19 at 3:11
• the Reference Level may depend on attenuators you select to protect the input diode mixer of the Spectrum Analyzer. In older SA, you had to find that setting, and perform add/subtract. On computerized SA, the display will show the Reference Level in dBm (0 dBm across 50 ohms is 0.223 volts rms, or 0.632 volts peakpeak) – analogsystemsrf Nov 13 '19 at 3:47
• The SA does not have a screen capture option. – Newbie Nov 13 '19 at 4:04
• the very top left reads "17 dBm", which is 50 milliWatts. – analogsystemsrf Nov 13 '19 at 5:21