# Measuring 150kHz, ~30V peak waveform with good resolution

I am trying to choose test equipment for a systems test application and I am running into a road block with one of the specs I need to meet.

The device features an H-bridge output similar to the circuit shown below, but instead of a motor we will use some sort of load power resistor (20 ohm, I believe). VBUS in our application is ~30V and the FETs are driven such that the output seen across the load is something close to a +/-30V peak, 150kHz waveform (approximately a square wave)

I am just interested in measuring the peak voltage of the waveform, but with relatively good accuracy. I need resolution down to about 1mV, ideally (I believe this would be 15 or 16-bit).

What sort of test equipment would be best suited to handle this measurement?

An oscilloscope seems to typically have poor resolution at its higher volts/div ranges (5V or 10V divisions in this case, likely), but would probably be preferable if affordable enough. A digitizer can have lower bandwidth and lower sample rate, but would seem to get me the resolution I need (PXI-4080 was one option I looked at, but the 0.1dB bandwidth is only 20kHz).

• Good question. Presumably you need the peak voltage on a cycle-by-cycle basis, so peak-detect circuits wouldn't be acceptable? Feb 4, 2020 at 15:55
• How much will the peak vary? Could you use an oscilloscope with the range set to the +/- deviation then the 30v offset subtracted? Feb 4, 2020 at 16:00
• Those MOSFETs on the right hand side are going to melt. Feb 4, 2020 at 16:14
• It would be helpful if you shed some light on why you're trying to make this measurement and what you're hoping to get out of it. Ideally, the load will either have +VBUS or -VBUS across it, depending how and when you take the measurement. In reality, there will be voltage drops across the MOSFETs, and potential ringing or oscillations due to parasitic inductance in traces and wires.
– BEE
Feb 4, 2020 at 16:15
• I believe that @Andyaka was commenting on the right hand side FETs are connected such that the body diodes will always be forward biased. They need to be flipped to match the left side.
– BEE
Feb 4, 2020 at 17:11