# Strange behaviour of a Wheatstone bridge

I am trying to do some strain measurements. I am using a strain gauge in a Wheatstone bridge configuration. The connection is shown in the image below:

simulate this circuit – Schematic created using CircuitLab

I am using an amplifier and some circuitry to amplify and read the Vo signal. SG1 is a strain gauge. R3 is set at resistance of 120 ohm. The voltage across SG1 and R2 (or) R3 and R4 is 2.6 V. As all the resistors are of equal value the voltage across each resistor should be 1.3V. When I turn on the Vex the voltage across R3 and R4 is 1.3 V each initially. After sometime I notice that the voltage across R3 goes up to 1.47 and the voltage across R4 is 1.1 roughly. The voltage across R3 was increasing like 1 mV/s. I tried to vary the potentiometer to balance the bridge, but the voltage was fixed across R3 and did not decrease.

Can anyone explain why does this happen and how to get rid of this?

I'm using through hole metal film resistors and soldered my connection on a prototype PCB.

• Voltage supply and bridge voltage are probably not constant. And current through SG1 and R3 probably also too "high", heating all resistors ... Commented Jul 22, 2022 at 11:14
• Is the 4.2V by any chance a lithium ion battery? Commented Jul 22, 2022 at 16:20
• The supply voltage Vex is provided by HX711 load cell amplifier. The HX711 is powered by Arduino 5V. HX711 has a voltage regulator that provides constant 4.2 V(Vex). I am using the R0 to minimise the power consumption/heating of resistors at bridge. The min common mode input of HX711 is 1.2 V, so providing bridge voltage near 2.5 V. That's the reason for choosing 75 ohm for R0. Commented Jul 23, 2022 at 0:23