# Why would my true RMS multimeter give me a different AC voltage reading than my scope at frequencies above 2 kHz?

Here’s the circuit and my problem. My function generator is simply parallel connected to a multimeter and my scope. No load is connected and I am not using a scope probe, just a BNC connector with alligator clips. Although I did try it with an 8 ohm precision resistor load connected in parallel to the scope, multimeter and function generator as well, but with no difference.

I was just checking to see if the reading on my scope for AC voltage is the same as the AC voltage reading on my multimeter. My multimeter is true RMS. The scope is calibrated and set to read AC RMS voltage. The scope has a 1 megohm input impedance. The function generator has a 50 ohm output impedance. I would think that the scope’s input impedance is transparent to the function generator and multimeter because it’s so high - so I don't believe that is the problem. I don’t think it’s reactance either. What’s weird is I can get the multimeter and scope to agree on AC RMS voltages between 500 Hz and about 2 kHz. Above 2 kHz the AC RMS voltage on the scope goes way below the AC RMS voltage on the multimeter, which would lead me to think that reactance is playing a part of this. But the scope only has 13 pF of input capacitance - so I looked at the capacitive reactance based on the frequencies and voltage I am using and I don't see that the readings should differ, so I don't think that's the problem either.

I set the function generator to put out 2 V of AC (as read by my multimeter) at 500 Hz (as entered on the function generator). Then I read the AC VRMS voltage on the scope and it matches the multimeter up to about 2 kHz. If the multimeter reads 2 V AC RMS then the scope reads 2 V AC RMS. Above 2 kHz, the multimeter always reads higher voltages than does the scope but never the same or by the same amount. Same thing happens if I crank up the amplitude of the function generator. The readings are always different above 2 kHz but never by the same amount. So I tried a different multimeter and get the same results.

Shouldn’t the multimeter and scope give me the same AC RMS voltage readings (within 10%) regardless of frequency? What is going on?

• The RMS to DC circuit in the multimeter may not work above 2 kHz. They do have bandwidth limitations. Does your meter spec sheet give any information on the accuracy vs frequency? Commented Aug 9, 2023 at 22:38
• Be certain to adjust the horizontal display so that only 1 or 2 cycles are displayed. As more cycles are displayed the resolution on the time measurement decreases. Commented Aug 9, 2023 at 22:38
• Most 50-ohm function generators I've seen are fairly careful to emit constant-amplitude waveforms regardless of frequency. So I'd be suspicious of any amplitude-measuring instrument that displays frequency-dependent readings. Commented Aug 9, 2023 at 23:13